The microhotplate, used as a basis for the microcombustor, is fabricated by through-wafer silicon etching. It consists of a silicon nitride membrane suspended from a frame of Si using either the Bosch etching or KOH etching to release the membrane, with no discernable operational differences between the completed devices made by either method.
The microhotplate, used as a basis for the microcombustor, is fabricated by through-wafer silicon etching. It consists of a silicon nitride membrane suspended from a frame of Si.
Schematic Cross Section
Either Bosch etching or KOH etching can be used to release the membrane, with no discernable operational differences between the completed devices made by either method. Due to their thermal sensitivity, typically better than 0.4 mW/°C, microhotplates have been used in a number of other applications including flow sensing, gas thermal conductivity detection, infrared bolometry, and conductometric gas sensing; typically using tin oxide as a gas sensitive layer for the latter.
|SEM of Microhotplate, front sid||SEM of Microhotplate, etch sid|
The microcombustor, consisting of a catalytic film deposited on the surface of a microhotplate, allows for sustained combustion on the microscale. This micromachined design has low heat capacity and thermal conductivity, making it ideal for heating catalysts placed on its surface. The catalytic materials provide a natural surface-based method for flame ignition and stabilization and are deposited using a micropen system, which allows precise and repeatable placement of the materials.
Sandia National Laboratories has been working on two gas sensors based on a microcombustor platform. While both sensors are in their early stages of testing and development, they exhibit good detection limits and signal quality. In addition to testing the sensors themselves, headway has also been made in understanding the combustion characteristics of hydrocarbons when exposed to a variety of catalysts. The two sensor technologies are a micro flame ionization detector (microFID) and a calorimetric gas sensor. Both are utilized in the determination of fuel mixture, though the microFID has a demonstrated analyte set that expands beyond hydrocarbon detection.
Cross-sectional Schematic of MicroFID
For additional information or questions, please email us at MGA@sandia.gov